Table of Contents
The manual method of rainfall measurement is the simplest method and includes the measurement of rainfall by means of an instrument which is known as a rain gauge.
The rain gauge works by sampling the incidence of rainfall at a specific point basically through an orifice of a known area.
There are 2 types of rain gauges that are explained below:
2. Types of Rain Gauge
1. Non-Recording Rain Gauge
Non-recording type rain gauge is the simple form of rain gauge that essentially collects the rainfall but does not record the quantity of the collected rainfall.
The non-recording rain gauge generally consists of a circular collecting area of about 12.7cm in diameter.
The circular collecting area is connected to a funnel.
The objective of providing the funnel is to discharge the collected rainwater to the receiving vessel.
The rim of the circular collector must be set at a horizontal plane at a height of about 30.5cm above the ground level.
The arrangement of the circular collector and the discharging funnel is kept inside a metallic container.
The rainfall is allowed to accumulate in the vessel and is duly measured by a graduated measuring jar.
The measuring jar has an accuracy of about 0.1mm.
The rainfall is measured in mm or cm as the depth of water.
The most commonly used non-recording rain gauge is the Symon’s rain gauge.
Figure: Non-recording Rain Gauge (Symon’s rain gauge)
2. Recording Rain Gauge
Recording Rain Gauge delivers a permanent automatic record of the rainfall.
This rain gauge is also commonly referred to as the Integrating rain gauge as it records the cumulative rainfall.
The recording rain gauge is arranged in such a way that the total amount of rainfall recorded is directly recorded on graph paper.
Such rain gauge gives the mass curve of rainfall i.e. plot of cumulative rainfall vs time.
The recording rain gauge also gives the duration of the rainfall as well as the intensity of the rainfall at any given time.
The recording rain gauges can be further classified into the following types:
a. Tipping Bucket Type Rain Gauge:
It is the type of rain gauge that consists of a pair of bucket arrangement that is arranged in such a way that when 0.25 mm of rain falls in one bucket, it tips automatically bringing the other bucket in position.
The rain collected from the tipped bucket is then collected in a can.
The tipping mechanism of the bucket actuates an electrically driven pen so as to trace the plot on the graph paper.
The graph is mounted on a clock-driven drum. The collected water is then measured to determine the quantity of rainfall.
The diameter of the tunnel which collects rainwater in the tipping bucket rain gauge is 300mm(30cm).
Figure: Tipping Bucket Rain Gauge
b. Weighing Bucket Type Rain Gauge:
The weighing bucket-type rain gauge consists of an arrangement of a bucket mounted on a weighing scale.
The weight of the bucket and its counterparts are duly recorded on a clock-driven chart.
Such rain gauge delivers the mass curve of rainfall i.e plot of the cumulative rainfall vs time.
Figure: Weighing Bucket Rain Gauge
c. Floating Type Rain Gauge:
It is also commonly referred to as the natural siphon type rain gauge.
This type of rain gauge consists of a funnel-shaped collector that leads to the floating chamber.
As the rainfall gets accumulated on the collector, the float subsequently rises and the pen attached to the float records the rainfall on a rotating drum through a clockwise mechanism.
The pen works through a lever system.
Figure: Floating Types of Rain Gauge
3. Factors To Be Considered During Selection of Rain Gauge Stations
Some of the factors that must be considered for the selection of the rain gauge stations are as follows:
1. The station must lie on a level and open ground so that the gauge can represent a horizontal catch surface. The rain gauge must not be surrounded by any object at least up to 30m.
2. The instrument must be sufficiently high so as to prevent splashing, flooding, etc.
3. The instrument must be placed near the ground as far as possible. This is done to minimize the wind effect.
|Read More: Mortar and Its types|
Educational Platform Under Naba Buddha Institute